Currently we have an X7-2T philips TOE probe in for repair. The probe has had several knocks to the head of the probe. The image quality is severely impacted. Check your probe today for any similar damage
You may also find damage to the face of the lens. This can look like this:
Damage like this is a sterilization hazard and also can effect imaging. We can repair this damage at an affordable price. Check your probes and contact ProbeLogic for friendly, helpful advice.